Puts the device into standby locally
and at the Information Center.
Displays purchased/enabled product
Displays available monitors and
IntelliVue Cableless Measurements
to assign to via the short-range radio.
Prints the pre-configured report as
designated at the Information Center.
View up to 24 hours of tabular trend
Determines time period that the
display remains active after user
Locks/Unlocks the display.
Selects either Monitoring, Demo,
Config or Service modes.
The Alarm Area of the MX40 displays
physiological alarms and technical alarms.
A multiple alarm indicator (down arrow) is
displayed when multiple alarm conditions
A check mark in front of the alarm text
signifies that the alarm has been
acknowledged by touching the Silence
Alarm Indicators display in the Patient
Information Area in place of the time clock
when alarm/INOP conditions are present
but have not been acknowledged.
Touching the Alarms Area displays a list of
all active alarms.
The alarms paused icon communicates
whether the alarm system is on/off.
Local Alarm Audio is off when the alarm