The equipment returns to the previous operating mode within 10 seconds
without loss of any stored data. Measurement accuracy may be temporarily
decreased while performing electrosurgery or defibrillation. This does not
affect patient or equipment safety. Do not expose the equipment to x-ray or
strong magnetic fields (MRI).
After power interruption, an ECG wave will be shown on the display after
30 seconds maximum.
150 kHz to 80 MHz
80 MHz to 800 MHz
Safety Standards & Specifications
800 MHz to 2.5 GHz