Device Assignment at the Information Center
You can assign an MX40 to a patient monitor at the Information Center. The
data from the MX40 automatically displays as a permanent overview
session in the Telemetry Data window on the patient monitor.
At the Information Center the MX40 data and the patient monitor data are
integrated in the patient sector.
All data presented in the Telemetry Data window are delayed for several
seconds. If you need realtime data, e.g. for defibrillation, always use the
ECG from the patient monitor.
Device Assignment at the MX40
To assign an MX40 to an IntelliVue Cableless
The MX40 will attempt to complete the assignment for a 2-minute period. If
assignment fails or the MX40 is no longer in range of the other device, the
short-range radio turns off to save battery power. Repeat the procedure
above to retry or resume the assignment. You may also need to restart the
short-range radio at the cableless measurement device if it has entered its
power save mode. Touch and hold the left button on the device to restart
the short-range radio.
To assign an MX40 to an IntelliVue Patient Monitor:
Monitoring with other Assigned Devices
Press the SmartKey button.
Press the Add/Remove SmartKey.
From the Add to menu, select the desired device and press Confirm.
Press the Smartkey button.
Press the Add/Remove Smartkey. The measurement selection key on
the monitor will change to show the "assign telemetry" icon
In the Assign Telemetry Device menu, select the correct equipment label
for the device.