1, 2, 3, 4,
5, 6, 7, 8
CONFIGURATION, SETUP, AND ADVANCED MODE
The FR2+ comes with a factory default configuration designed to meet the
needs of most users. If desired, your Medical Director can revise the setup.
Some setup parameters govern specific features that are not related to the
patient care protocol, some are used to define the automatic patient care
protocol used by the FR2+, and some provide options for manual override of
NOTE: The configuration features discussed in this manual are for FR2
software version 1.7. Certain functions of this software are new or differ
from previous software versions. Contact Philips for information on how to
upgrade your FR2 or FR2+ to the latest software version.
In addition, the configuration settings information provided in Edition 5 or
earlier of the Training & Administration Pack Instructions for Use is
superseded by the information in this chapter. Other directions for use of
the Training & Administration Pack provided in its Instructions for Use
The following table presents parameters that do not affect the treatment
Sets volume of the FR2+'s speaker. 1 is lowest; 8 highest. The speaker is
used for voice prompts and the armed-for-shock tone.
Enables or disables the audio recording during incident. Voice recording
requires use of a data card.
Enables (ON) or disables (OFF) ECG display on the screen of the M3860A
only. FR2+ rhythm analysis does not require ECG display to be on. (ECG
display cannot be changed from the default, OFF, for M3861A.)