For proper grounding of the 854 ATG, install a copper strip under one of
Figure 4 Example of flange ground
Also in case of tanks with cathodic protection, the 854 ATG must be grounded to the tank.
Cathodic protection can be maintained by isolating the cable screen/armouring or conduit.
the flange bolts for each flange.
Place shark rings between flange and strip (figure 4).
Installation guide 854 ATG