Show ST Limits
Hide ST Limits
You can use the pop-up keys that open with the
All Lim. Narrow
These pop-up keys are not available in the window for changing individual alarm limits which you
access by selecting the measurement label in the
Changing Alarm Limits
To change individual measurement alarm limits using the measurement's Setup Menu,
In the measurement's setup menu, select the alarm limit you want to change. This calls up a list of
available values for the alarm limit.
Select a value from the list to adjust the alarm limit.
For example, to change the alarm limits for SpO
Touch the high limit on the alarm limits menu button.
Choose the high alarm limit from the pop-up list that opens.
Touch the low limit on the alarm limits menu button.
Choose the low alarm limit from the pop-up list that opens.
Alternatively, you can use the keys in the measurement alarm limits window, which you access by
selecting the measurement label in the
to expand the list of ST leads and view the currently set alarm limits. Selecting
hides the list again.
All Lim. Wide
to set narrow or wide alarm AutoLimits for all measurements.
Graphic view of current yellow
and red alarm limits and
indicates the measurement is
Measurement labels, with alarms
off symbol where appropriate
Set narrow or wide alarm
AutoLimits for all
window to perform common tasks: