1 Basic Operation
Depending on your configuration, the monitor will either
take no action, you must enter the
selection key) and resolve the conflict
All the currently available measurements are depicted in the
measurement labels causing a label conflict are shown in red. If a measurement is connected but
currently unavailable, for example, because it was deactivated due to a label conflict, that measurement
is shown "grayed-out". If an MMS Extension is not available, for example if monitor is running on
battery power and not an external power source, the MMS Extension is not displayed.
Unavailable measurement shown grayed-out
An MMS Extension is only shown in the
connected to AC mains via the external power supply (M8023A) or is attached to the Battery
To resolve a label conflict:
Press the SmartKeys key and
Or select the
to display the
Select the label to be corrected.
Use the measurement selection keys to resolve the conflict. Select either:
window automatically for you to resolve the conflict
Measurement selection key
: to assign a different label to the conflicting label.
window (using the measurement
window when the monitor is