Radiofrequency Radiation Exposure Information
For body worn operation, this equipment has been tested and meets the FCC RF exposure guidelines
when used with the accessories supplied or those approved for use with this product. Use of other
accessories may not ensure compliance with FCC RF exposure guidelines.
The MP2 and 865244 Remote Control Including the Short Range
FCC and Industry Canada Radio Compliance: This device complies with Part 15 of the FCC Rules and
RSS-210 of Industry Canada. Operation is subject to the following two conditions: (1) this device may
not cause harmful interference, and (2) this device must accept any interference received, including
interference that may cause undesired operation. Any changes or modifications to this equipment not
expressly approved by Philips Medical Systems may cause harmful radiofrequency interference and
void your authority to operate this equipment.
The radio component contained in this device is in compliance with the essential requirements
and other relevant provisions of Council Directive 1999/5/EC (Radio Equipment and
Telecommunications Terminal Equipment Directive)
In addition the product complies with: ETSI EN 300 328; AS/NZS 4268; ARIB STD-T66.
Class 1 radio equipment.
To obtain a copy of the original Declaration of Conformity, please contact Philips at the address given
in the "Manufacturer's Information" section of these Instructions for Use
Out-Of-Hospital Transport - Standards Compliance
The MP2 patient monitor with measurements and interfaces other than those listed below cannot be
used for patient transport outside of the hospital environment.
The MP2 patient monitor, with the following measurements and interfaces:
ECG/Respiration, NBP, SpO
LAN, internal Battery
can be used in a transport environment such as a road ambulance, airplane or helicopter. For this
purpose the monitor fulfils the following additional mechanical, EMC and environmental
Shock Tests according to IEC TR 60721-4-7, Class 7M3. Test procedure according to IEC/EN
60068-2-27 (peak acceleration up to 100g).
Random Vibration according to IEC TR 60721-4-7, Class 7M3. Test procedure according to
IEC/EN 60068-2-64 (RMS acceleration 5g).
Sinusoidal Vibration according to IEC TR 60721-4-7, Class 7M3. Test procedure according to
IEC/EN 60068-2-6 (acceleration up to amplitude 2g).
Bump Test according to IEC/EN 60068-2-29 (peak acceleration 15 g, 1000 bumps).
Free Fall Test according to EN1789 (covers also IEC TR 60721-4-7 and Class 7M3). Test
procedure according to EN 60068-2-32 (height 0.75 m).
Specification for degrees of protection provided by enclosures according to IEC/EN 60529: IP
, Pressure, Temperature, CO
(only Mainstream Sensor