This chapter lists Philips-approved accessories for use with patient monitors as described in this
document. You can order parts and accessories from Philips supplies at www.medical.philips.com or
consult your local Philips representative for details. Some accessories may not be available in all
All accessories listed in this chapter comply, in combination with the patient monitor, with the
requirements of IEC 60601-1-2.
Reuse: Never reuse disposable transducers, sensors, accessories and so forth that are intended for
single use, or single patient use only. Reuse may compromise device functionality and system
performance and cause a potential hazard.
Philips' approval: Use only Philips-approved accessories. Using non-Philips-approved accessories
may compromise device functionality and system performance and cause a potential hazard.
Packaging: Do not use a sterilized accessory if its packaging is damaged.
Correct Use: Always follow the instructions for use supplied with the accessories.
against electric shocks (particularly regarding allowable leakage currents), and are defibrillator proof.
The following cables may not all be available in all countries. Please check availability with your local
For use with:
3-Electrode Cable Set
3-Electrode Cable Set, OR
5-Electrode Cable Set
5-Electrode Cable Set, OR
This symbol indicates that the cables and accessories are designed to have special protection