Enabling Sequence Mode and Setting Up The
Up to four measurement cycles can be set up which will run consecutively. For each cycle you can
set the number of measurements and the interval between them. If you want to run less than four
cycles in a sequence, you can set the number of measurements for one or more cycles to
Select each sequence in turn and select the number of measurements and the time interval between
To have measurements continue after the sequence, set the number of measurements for your last
Be aware that, if none of the cycles are set to
measurement of the cycle.
When the NBP measurement mode is set to
Choosing the NBP Alarm Source
You can monitor for alarm conditions in systolic, diastolic and mean pressure, either singly or in
parallel. Only one alarm is given, with the priority of mean, systolic, diastolic.
Sys & Dia
Dia & Mean
Sys & Mean
is not selected as alarm source (
a mean value, mean alarms will nevertheless be announced using the most recent mean alarm limits.
Check that the mean alarm limits are appropriate for the patient, even when not using mean as the
alarm source. When no value can be derived an
to open the
and this cycle will run indefinitely.
Pressure value monitored
systolic and diastolic in parallel
diastolic and mean in parallel
systolic and mean in parallel
all three pressures in parallel
from the pop-up menu.
, NBP monitoring will end after the last
, the repetition time for
and choose from:
Sys & Dia
selected), but the monitor can only derive
NBP Measure Failed
11 Monitoring NBP
mode cannot be
INOP will be displayed.