1 Basic Operation
MMS Extension M3014A attached to the MP2
The MMS extensions connect to the monitor and use the monitor's settings. Trend data and
measurement settings from the measurements in the extensions are stored in the monitor.
Measurements from an MMS extension are only available when the extension is connected to the
monitor, and the monitor is running on external power. This is the case when the monitor is
running on AC mains via the external power supply (M8023A) or is attached to the Battery
Extension (865297). Measurements from an MMS extension connected to the monitor are not
available when the monitor is running on its own battery power.
Any measurements on an MMS extension that conflict with those in the monitor cannot be used.
For example, only one CO
The Cardiac Output and Continuous Cardiac Output option of the MMS extensions is not
available for the MP2 monitor.
To separate an extension from the monitor, press the release lever and push the extension forward.
measurement is supported.