Analysis results included in reports and additional report selections with ST Map.
Remote operation of the 12-Lead Export function at the IntelliVue Information Center.
Remote operation of the 12-Lead Lock/Unlock function at the IntelliVue Information Center.
New filter settings that are used as a default for future 12-Lead captures.
New "End Afib" and "End Irregular HR" Alarms
The end of an atrial fibrillation or irregular HR phase is announced with an
Information Line Changes
Patient Category is now displayed as a symbol in front of the patient name and cannot be changed
from the Information line. Paced Mode is no longer displayed in the Information line; it appears next
to the HR numeric.
Depending on your configuration, you will see the current Profile name, and/or the Screen name in
the Information line. The Profiles menu has been simplified in Monitoring and Configuration Mode.
Sepsis Protocol Configurability
The SSC Sepsis protocol can now be configured, to customize it for the specific treatment measures
used in your facility. This can include changes to limits for measured or manually entered values,
changes to the Severe Sepsis Screening criteria, and changes to the recommendations in the Sepsis
Resuscitation Bundle and the Sepsis Management Bundle.
Integrated Pulmonary Index for Microstream CO2
An Integrated Pulmonary Index (IPI) numeric is provided, that is an indication of the patient's overall
ventilatory status based on four measurement parameters: etCO
Cardiac Output (transpulmonary method) - Changes to Calculation
Basis for Extravascular Lung Water Index and Preload Indexes
EVLWI is now calculated using Ideal Body Weight (IBW). ITBVI and GEDVI are calculated using
Ideal Body Surface Area (IBSA). IBW and IBSA are based on the patient category, gender and height
that are entered for the patient.
Improved Visibility of Gridlines in Overlapping Waves
Visibility of the gridlines for overlapping waves has been improved, and you can now configure
gridlines to be shown in white, if you prefer, in a thin or thick style.
Timeout for "Aged" Numerics
NBP and SpO
disappear from the screen after a set time. This avoids older numerics being misinterpreted as current
data. The time can be set individually for SpO
alarm. The delay time before the alarm is announced can be configured.
numerics from intermittent measurements can be configured to be grayed out or to
, awRR, pulse rate and SpO
and NBP in Configuration mode.
2 What's New?
** End AFIB