Unless otherwise specified, this information is valid for SpO
measured using the M3001A and
M3002A Multi-measurement modules and the M1020B measurement module. The SpO
Specifications in this section apply to devices with Philips SpO
technology. For SpO
Specifications valid for other SpO
technologies, refer to the instructions for use provided with these
Complies with ISO 9919:2005 / EN ISO 9919:2009 (except alarm system; alarm system complies with
Measurement Validation: The SpO
accuracy has been validated in human studies against arterial
blood sample reference measured with a CO-oximeter. Pulse oximeter measurements are statistically
distributed, only about two-thirds of the measurements can be expected to fall within the specified
accuracy compared to CO-oximeter measurements.
Display Update Period: Typical: 2 seconds, Maximum: 30 seconds. Maximum with NBP INOP
suppression on: 60 seconds.