X2 Performance Specifications
X2 (M3002A) Performance Specifications
Real Time Clock
(with new, fully charged battery at
On/Standby / Error
System alarm delay
Extended alarm pause
if powered by AC: infinite
without power: at least 8 hours
Active settings, trends, patient data, realtime reports, events,
<12 W average
<30 W while battery is loading
36 to 60 V DC floating
Basic monitoring configuration: 2.5 hours
(Brightness set to optimum, ECG/Resp, SpO
in use, NBP measurement every 15 minutes)
When X2 is off: 2 hours
When X2 is in use, and connected to a monitor, without
extensions: 12 hours approx.
When X2 is in use, and connected to the external power supply
(M8023A), without extensions: 12 hours approx.
red or yellow LED with crossed-out alarm symbol
red/yellow/light blue (cyan) LED
green / red LED
yellow (charging)/red blinking (empty) LED
Audible feedback for user input
QRS tone, or SpO
4 different alarm sounds
12 or 16 numerics @ 12 sec, 1 minute, 5 minute resolution.
Multiple choices of number of numerics, resolution and
duration depending on trend option and application area.
For neonatal, you can choose between 12 and 16 numerics.
less than 4 seconds
1,2,3 minutes or infinite, depending on configuration
5 or 10 minutes
all alarms / inops, main alarms on/off, alarm silence and time
from: January 1, 1997, 00:00 to: December 31, 2080, 23:59
<4 seconds per day (typically)
infinite if powered by host monitor or external power supply;
otherwise at least 48 hours