Never carry a monitor by the handle when equipment is attached below it with the Quick Mount.
To transport such equipment combinations, always support the equipment from below.
Only use the bed hanger for the MX400/450/500/550 on beds with a support below the rail, to
ensure that the monitor does not tilt.
Repositioning a Monitor on the Mounting Arm
Attempts to reposition the monitor should only be performed as described in the mounting hardware
manufacturer's user documentation. If the mounting arm has a locking mechanism, ensure the locking
mechanism is unlocked before attempting to reposition the monitor. Never pull on the monitor to tilt,
swivel or otherwise reposition it - always hold on to the mounting hardware itself.
Not adhering to these instructions when repositioning the monitor can cause damage to the monitor.
In extreme cases, when force is applied to the monitor, it can result in the monitor falling from the
If, at any time, the monitor appears to be loose or insecurely mounted, contact your service personnel.
Altitude affects tcGas and CO
Monitor Safety Specifications
The monitors, together with the Multi-Measurement Modules (M3001A/M3002A), the Flexible
Module Racks FMS-8 (M8048A) and FMS-4 (865243), and all modules and MMS extensions, comply
with the Medical Device Directive 93/42/EEC.
Covidien BISx, Covidien BIS Engine and Covidien DSC comply with the Medical Device Directive
93/42/EEC of 14 June 1993.
In addition, the product complies with:
IEC 60601-1:1988 + A1:1991 + A2:1995; EN 60601-1:1990 + A1:1993 + A2:1995; UL 60601-1:2003;
CAN/CSA C22.2#601.1-M90 +Suppl. No. 1-94 + Am. 2; IEC 60601-1-1:2000; EN 60601-1-1:2001;
IEC 60601-1-2:2001 + A1:2004; EN 60601-1-2:2001 + A1:2006.
The possibility of hazards arising from software errors was minimized in compliance with
ISO 14971:2007, EN 60601-1-4:1996 + A1:1999 and IEC 60601-1-4:1996 + A1:1999.
Classification (according to IEC 60601-1): Class 1, Type CF, Continuous Operation. The BIS and
NMT measurement use a Type BF applied part.
measurements. The monitor must be configured at installation to the