Measurement selection key with question marks indicating a label conflict.
Depending on your configuration, the monitor will either
resolve the conflict automatically, by assigning a new, generic label to the most recently connected
conflicting label (e.g. a second FAP label could be changed to ABP)
take no action, you must enter the
All the currently available measurement devices, for example MMSs, MMS extensions, measurement
modules, Gas analyzers, devices connected to a host monitor and manually entered measurements
), are depicted in the
Any measurement labels causing a label conflict are shown in red. If a measurement device is
connected but currently unavailable, for example, because it was deactivated due to a label conflict, the
device is shown "grayed-out".
When an X2 is connected to a host monitor, the measurement selection window looks like this:
X2 connected to a host monitor
When an X2 with an active measurement, say SpO
measurement already active, the SpO
INOP is displayed. The measurement can only be reactivated if the X2 is disconnected
from the host monitor. The label conflict can be resolved on the host monitor like any other label
window automatically for you to resolve the conflict
De-activated devices are grayed-out (here
the Temp measurement on the left)
, is connected to a host monitor with the same
measurement on the X2 is deactivated and the
window and resolve the conflict.