Installation Safety Information
If multiple instruments are connected to a patient, the sum of the leakage currents may exceed the
limits given in IEC/EN 60601-1, IEC 60601-1-1, UL 60601-1. Consult your service personnel.
The monitor must be grounded during operation. If a three-wire receptacle is not available, consult the
hospital electrician. Never use a three-wire to two-wire adapter.
If the monitor is used in internal examinations on the heart or brain, ensure that the room incorporates an
equipotential grounding system to which the monitor has a separate connection.
Combinations of medical equipment with non-medical equipment must comply with IEC 60601-1-1. Never
use a multiple portable socket-outlet or extension cord when combining equipment unless the socket outlet is
supplied specifically for use with that equipment.
The monitor uses double fusing (line and neutral).
All network cables must be unshielded.
The actual placement of boards and configuration of connections for your monitor depends on how
your hardware has been configured. See the symbols table ("Symbols" on page 445) to see which
symbols are used to mark the connections.
Connect only medical devices to the ECG output connector socket.