Care and Cleaning
Use only the Philips-approved substances and methods listed in this chapter to clean or disinfect your
equipment. Warranty does not cover damage caused by using unapproved substances or methods.
Philips makes no claims regarding the efficacy of the listed chemicals or methods as a means for
controlling infection. Consult your hospital's Infection Control Officer or Epidemiologist. For
comprehensive details on cleaning agents and their efficacy refer to "Guidelines for Prevention of
Transmission of Human Immunodeficiency Virus and Hepatitis B Virus to Health Care and Public-
Safety Workers" issued by the U.S. Department of Health and Human Services, Public Health Service,
Centers for Disease Control, Atlanta, Georgia, February 1989. See also any local policies that apply
within your hospital, and country.
Keep your monitor, modules, Multi-Measurement Module, MMS extensions, Flexible Module Rack,
cables and accessories free of dust and dirt.
After cleaning and disinfection, check the equipment carefully. Do not use if you see signs of
deterioration or damage. If you need to return any equipment to Philips, decontaminate it first.
Observe the following general precautions:
Always dilute cleaning agents according to the manufacturer's instructions or use lowest possible
Do not allow liquid to enter the case.
Do not immerse any part of the equipment or any accessories in liquid.
Do not pour liquid onto the system.
Never use abrasive material (such as steel wool or silver polish).
Never use bleach except in an approved product listed in this chapter.
If you spill liquid on the equipment, battery, or accessories, or they are accidentally immersed in liquid,
contact your service personnel or Philips service engineer. Do not operate the equipment before it has
been tested and approved for further use.
The general care and cleaning information given here meets the requirements of Covidien llc for their
BIS measurement devices.