29 Printing Patient Reports
When an X2 is connected to a host monitor, reports requested or stored on the X2 can be printed via
the host monitor. Reports requested on the X2 when connected to a host monitor will not show any
patient alarms or INOPs. In order to see alarms, reports should be requested on the host monitor.
Starting the Different Types of Report
Vital Signs Report
Depends on selected trend group, data
resolution, and period.
Graphic Trends Report
Graphic or tabular Event Review
EEG Waves, numerics, CSAs, montage
information and impedance values
Depends on format selected.
Captured ECG waves, measurement values, ST
Map (optional) and analysis results from the
Information Center (PIIC iX only)
ST Map Report
ST data in a transversal and/or horizontal multi-
Cardiac Output Report
Trial curves and numerics
Wedge numerics and reference wave
Hemodynamic, Oxygenation, or Ventilation
Drug Calculator Report
Alarm Limits Report
Graphic and numeric report of all current alarm
Patient data and numerics, and either:
all displayed waves OR all measured waves OR
all measured RT waves OR all measured HiRes
waves, OR oxyCRG waves (RT waves: ECG
Primary lead, Pleth, Resp; HiRes waves: btbHR,
Individual Loops diagram with timestamp,
currently measured numerics
Central Report A/B/C
These are reports configured at the Information
Center. Monitors connected via the IntelliVue
Instrument Telemetry wireless network can have
these reports available.
, HiRes Resp)
How to start printing reports
In the Vital Signs window, select Print/ Record, then
In the Graphical Trends window, select Print.
In the Event Episode or Event Review window, select
the pop-up key Print/ Record, then select Print
Review or Print Episode.
In the Setup EEG menu, select Print Report.
Select the ECG Report A or ECG Report B SmartKeys,
if configured, or the Print Reports SmartKey followed
by ECG Report A or ECG Report B.
In an active 12-Lead application window, select the Print
Report pop-up key
In the ST Map (Current) or ST Map (Trend) window,
In the Cardiac Output (Right Heart) or Cardiac
Output (Transpulmonary) window, select the pop-up
key Print/ Record.
In the Wedge window, select the Print Wedge pop-up
In the Calculations windows, select the Print/ Record
In the Titration Table window, select the Print
Titr.Tbl pop-up key.
In the Drip Table window, select the Print Drip Tbl
In the Alarm Limits window, select Print Limits.
Select the Realtime Report SmartKey, if configured.
In the Loops window, select Print Loop.
In the Main Setup select Reports then Central
Report A, Central Report B or Central Report C.