22 Using a Telemetry Device and a Monitor (PIIC only)
Enter here the equipment label of the telemetry device to be paired.
Pairing at the monitor is only possible when the monitor already has a connection to the Information
Center and the Information Center software version allows pairing at the monitor.
Own Patient Overview Window
The measurements from the telemetry device are shown in the own patient overview window on the
monitor. See "Own Patient Overview Window" on page 122
Direct Connection - Automatic Pairing
The following direct connections are possible:
an X2, declared as a telemetry device, and connected with an MSL cable to a monitor.
When ECG is being measured with a telemetry device directly connected to the monitor, there will
be no ECG signal available at the ECG analog output or ECG Sync Pulse output and no
synchronization marks on the ECG wave. A
When a telemetry device is connected to the monitor, arrhythmia relearning is initiated, and again
when the telemetry device is disconnected.
Controls on the Telemetry Device (e.g. nurse call) will be inactive when the device is directly
connected to the monitor except in the case when the monitor has no network connection and
data are transferred via the telemetry device.
Pairing an X2 With a Telemetry Interface to a Host Monitor
An X2 Multi-measurement module using an IntelliVue Instrument Telemetry interface will be
automatically paired when it is connected to a host monitor and it has been declared as a "telemetry
device" at the Information Center.
When the X2 is disconnected from the host monitor it will remain paired and the measurement data
will appear at the host monitor in the Telemetry Data window. After the X2 is disconnected from the
monitor it may take over 15 seconds until their data is displayed at the Information Center.
Pairing of a host monitor and a telemetry device, with all the related functionality, is only possible
when the host monitor is using a wired LAN connection or a wireless LAN interface (but not using
the IntelliVue Instrument Telemetry interface).
The X2 can also be manually paired to a host monitor without a direct connection, as described in
"Indirect Connection - Manual Pairing" on page 307.
Refer also to "Use Models With Telemetry" on page 309 for further related use modes.
menu will appear with only one entry
No ECG Out
message will appear in the ECG wave