X1 Connectors and Symbols
X2 Multi-Measurement Module (M3002A)
The X2 Multi-Measurement Module (MMS) can simultaneously monitor 3-, 5-, 6- or 10-lead ECG
(including arrhythmia and ST monitoring), respiration, SpO
temperature, or CO
The X2 has the added capability to operate as a stand-alone monitor, and can be powered by a
rechargeable battery. This makes it particularly suited to transport situations. When the X2 is
disconnected from the original host monitor, it continues to monitor the patient as a stand-alone
monitor running on battery power, eliminating the need for a separate transport monitor. When the
X2 is connected to a new host monitor, it resumes its role as MMS, ensuring fully continuous
monitoring. For details of using the X2 as a stand-alone monitor, refer to the IntelliVue X2
Instructions for Use.
When connected to a host monitor (
host, including that required for battery charging. The X2 can also be powered by AC mains when not
connected to a host monitor using the optionally available external power supply (M8023A). See the
IntelliVue X2 Instructions for Use for details.
. It has a color touchscreen display.
White ECG/Resp connector
Red NBP connector
Combined pressure (red) and temperature
(brown) connector - connect either invasive
pressure transducer or temperature probe.
You might have a version of the MMS that
does not have this connector.
NBP STAT key - starts NBP STAT series
Zero key - initiates a zero procedure for the
connected pressure transducer when
pressed and held for a second
NBP Start/Stop key - starts or stops NBP
Silence: acknowledges all active alarms by
switching off audible alarm indicators and
, NBP and either invasive pressure and
is indicated), the X2 takes power from the