Select each sequence in turn and select the number of measurements and the time interval between
To have measurements continue after the sequence, set the number of measurements for your last
Be aware that, if none of the cycles are set to
measurement of the cycle.
When the NBP measurement mode is set to
Choosing the NBP Alarm Source
You can monitor for alarm conditions in systolic, diastolic and mean pressure, either singly or in
parallel. Only one alarm is given, with the priority of mean, systolic, diastolic.
Sys & Dia
Dia & Mean
Sys & Mean
is not selected as alarm source (
derive a mean value, mean alarms will nevertheless be announced using the most recent mean alarm
limits. Check that the mean alarm limits are appropriate for the patient, even when not using mean as
the alarm source. When no value can be derived an
Switching Pulse from NBP On/Off
In the process of making the NBP measurement, a pulse value can be derived and displayed. The pulse
value is displayed together with the time the measurement was made. After one hour the value
becomes invalid. There are no alarms associated with pulse from NBP.
When the accelerated measurement is in use, no pulse value can be derived.
To switch the display of the pulse value on or off:
and this cycle will run indefinitely.
Pressure value monitored
systolic and diastolic in parallel
diastolic and mean in parallel
systolic and mean in parallel
all three pressures in parallel
, NBP monitoring will end after the last
, the repetition time for
and choose from:
Sys & Dia
selected), but the monitor can only
NBP Measure Failed
10 Monitoring NBP
mode cannot be
INOP will be displayed.