ST View Window
Window shows a current ST snippet and numeric with a baseline snippet and numeric.
The two snippets are in different colors, so that you can differentiate between them easily and see at a
glance which numerics belong to which snippet. In the upper left corner the current lead and the ISO/
J-Point mode for the current snippet and the baseline snippet is shown. On the right side all available
ST numerics are shown. The date and time of the baseline snippet is shown at the bottom of the
Window opens with the ST pop-up keys
and arrow keys for scrolling through the available leads.
To view the
window, select any snippet on the Screen.
ISO/J-point mode for current snippet
ISO/J-point mode for baseline snippet
1 mV calibration bar
Current ST snippet and ST baseline snippet overlapped
Timestamp of most recently stored baseline snippet
All available ST numerics for current snippet and baseline snippet
The ST point, J-point and ISO point can be hidden using the
STE View Window
window shows a current ST snippet and STE numeric with an ST baseline snippet. The
two snippets are in different colors, so that you can differentiate between them easily. In the upper left
corner the current lead is shown. On the right side all available STE numerics are shown. The date and
time of the ST baseline snippet is shown at the bottom of the window.
6 ECG, Arrhythmia, ST and QT Monitoring
(to select viewing
Adjust ST Points