5 Managing Patients and Equipment
Calculation data (HemoCalc
If the monitor is not battery-powered, you cannot monitor during transport.
Measurements from an MMS extension connected to an X2 are not available when the X2 is running
on its own battery power. They are only available when the X2 is powered by external power: when
connected to a host monitor, to the external power supply (M8023A) or to the Battery Extension
The following sections describe the transfer procedures with a PIIC iX, a PIIC or with a standalone
Data Exchange Between Information Centers
You can transfer demographic data and trend data from one IIC to another by selecting
the patient monitor. Trend data is not shared between Information Centers and monitors.
Transferring Patients with IntelliVue Information Center iX
There are several ways to transfer a patient from or to another bed. When a patient is transferred to the
destination bed, Information Center monitoring continues in the destination bed and equipment that is
not transferred with the patient is freed up.
Using the Transfer Function
To transfer a patient using the transfer function provided on the patient monitor by the Information
Select the patient name or the
transfer that patient to another bed (
window offers the possibility to transfer a patient from another bed using the patient search as
described in "Using "Find Patient" to Search for Patient Data" on page 107.
selected a unit, or if only one unit is configured, a list of free beds will appear.
Select a bed from the list.
You will be asked to confirm the transfer of the patient to this bed.
Stored in Monitor
window opens. If a patient is currently admitted, the window offers the possibility to
To Other Bed
, if more than one unit is configured, a list of units will appear. When you have
to complete the transfer.
Stored in X1 MMS and
text at the top of the monitor screen to open the
To Other Bed
). If no patient is currently admitted, the
Stored in X2 MMS and