If any abnormality arises during use, please check the following points
Unit does not turn
on when the START/
STOP button is
have been positioned
EE measurement error
Cuff has been placed
symbol shown on dis-
play or the blood pres-
sure value is displayed
Did you talk or move
Shaking of the arm with
the cuff on.
E1 error symbol shown
Air circuit abnormality.
Cuff tube may not be
plugged into monitor
E2 error symbol shown
exceeding 300 mmHg.
E3 error symbol shown
Note: If the unit still does not work, contact HoMedics Consumer Relations.
Under no circumstance should you disassemble or attempt to repair the
unit by yourself. Contact information for HoMedics Consumer Relations
Department can be found on the warranty page.
Replace them with
four new AAA
Re-insert the batteries
in the correct positions.
Wrap the cuff properly
so that it is positioned
Measure again. Keep
arm steady during
Check cuff connection.
Switch the unit off, then
Four 1.5V DC (AAA) batteries
Measurement Method: Oscillometric
Pressure: 40~280 millimeters mercury (mmHg)
Pulse: 40~199 beats/minute
Pressure: ±3 mmHg
Pulse: ±5% of reading
Automatic Pressure release valve
1 minute after last button operation
Temperature: 50°F~104°F (10°C ~40°C)
Humidity: 15% ~ 90% RH
Temperature: -4°F~158°F (-20°C~70°C)
Humidity: Less Than 90% RH
0.48 lbs (217.6 g) (without batteries)
Standard size cuff: 9"–13" (23-33 cm)
Large size cuff (not included):13"–17" (33-43 cm)
5.16"(L) x 3.98"(W) x 2.05"(H)
131.4mm(L) x 101mm(W) x 52mm(H)
Arm cuff with tube, instruction manual
Internally Powered equipment
Type BF Equipment
IPx0-No special protection against the
ingress of water
Not suitable for use in presence of flammable
anesthetic mixture with air, oxygen or
Continuous operation with short-time loading.
Note: These specifications are subject to change without notice.