You primarily monitor your patient's ECG waveform in Monitor View by turning the Therapy knob to
In Monitor View, you can review three waves simultaneously while monitoring all current vital sign
Selecting the Waveform
The HeartStart XL+ allows you to configure the lead displayed as the primary ECG lead in Wave
Sector 1 when the device is turned on. The factory default is Lead II. See
When you first turn the HeartStart XL+ on in Manual Defibrillation, Synchronized Cardioversion or
Monitor modes or switch into one of those modes, the default lead is displayed in Wave Sector 1. If the
default lead is not available or is of poor quality, the device automatically searches for the ECG lead with
the best quality and displays that new lead in Wave Sector 1.
You can also change the waveform for each sector during a patient event.
The ECG wave for Wave Sector 1 is selected through the Lead Select button
on page 23) or through the Displayed Waves menu. The waves for Wave Sectors 2 and 3 are
selected through the Displayed Waves menu only.
To select a waveform using the Displayed Waves menu:
Press the Menu Select button.
Using the Navigation buttons, select
Select the wave sector you want to modify and press the Menu Select button.
Select the new wave type you want and press the Menu Select button.
Monitor View Layout
25 Aug 2011 10:44 AM
and press the Menu Select button (see
4: ECG Monitoring
>= 25 Kg
Table 33 "Wave Settings"