NOTE: Once you have pressed the Sync button, you can leave the HeartStart XL+ unattended as Operational
Check completes its process. If you cancel the Operational Check before it completes all tasks, there is no
record of the check in the Operational Check Summary.
10 After the automated part of Operational Check concludes, an Operational Check Report is printed
"Printing Operational Check Results"
WARNING: When the shock button is pressed during Operational Check, energy is delivered through the
pads/paddles into the defibrillator test plug/test load or paddles pockets. Make sure that the test plug or
test load is connected or that the paddles are seated in the holders properly.
NOTE: It is important to complete all instructions listed on the Setup screen in order to successfully complete an
Operational Check. Approximately 10 seconds after entering Operational Check, a
prompt appears in the lower right corner of the display. This prompt allows you to continue with the
Operational Check if the device is not responding to actions you have taken during Setup. Selecting
Proceed As Is
Operational Check Setup Screen
25 Jan 2011 11:10pm
(Device Reference Number)
(Device Serial Number)
Last Operational Check: 23Jan2011 04:32PM Pass
1. Connect Therapy Cable (required)
2. Turn knob to 170 (Required)
while there are still required items listed in the setup instructions causes Operational
14: Operational and Shift Checks
on page 158).
Proceed As is
Proceed As Is