12: Data Management
Event Information (Continued)
Disarm Auto (reason)
Shock # (number) (energy) J
(impedance) (peak current)
Shock Aborted (impedance) Logged when a shock is initiated but aborted before a full shock dose is delivered.
Abnormal Shock Dose
No Shock Advised
Cannot Analyze ECG
Pacer Mode Demand/Fixed
Pacer Rate (value) ppm
Pacer Output (value) mA
Logged when the device disarms automatically. Reasons include:
Pads Off: Logged when the automatic disarm is caused by a bad connection between the
device and the patient.
Shock Equipment Malfunction: Logged when the device is unable to reach the selected
energy during charging.
Timeout: Logged when the device reaches its configured auto-disarm period.
Shock Aborted: Logged when the shock is aborted.
No Shock Advised: Logged when, in AED Mode, the algorithm determines the rhythm is
Device Off: Logged when the device is turned off while charged.
Leads Off: Logged in Synchronized Cardioversion Mode when a leads off condition is
detected in the synchronizing lead.
Pads/Paddle Type Unknown: Logged when, with the therapy cable connected, the device
detects a change in the paddles or pads type or if the therapy cable type identification is
Logged when a shock is delivered.
Logged when Sync is turned on or off and any subsequent change.
Logged when a shock is initiated and completed but the full dose is not delivered.
Logged when the algorithm begins analysis.
Logged when artifact has been detected.
Logged when the algorithm detects a shockable rhythm.
Logged when the algorithm detects a non-shockable rhythm.
Logged when the algorithm is unable to make a shock/no shock decision.
Logged when the device enters or exits a forced pause.
Logged when the device enters or exits a NSA pause.
Logged when the device enters or exits NSA monitoring.
Logged when the device enters or exits CPR pause.
Logged when pacing starts and when the mode is changed.
Logged when the Pacer Rate is changed.
Logged when the Pacer Output is changed.
Logged when pacing is paused.