Pacing alarms can be generated for the conditions shown in
alarm messages in the Pacer Bar. There are both audio and visual alerts. When you switch patient
categories, all parameter alarm limits change to the new patient category. These changes are retained
when you switch modes.
For patients that are ≥ 25kg or ≥ 8 years old, use Adult patient category.
For patients <25kg or < 8 years old, use Infant/Child patient category.
For more information on alarms, see
Pads Cable Off.
Pacer Output Low
NOTE: Once the reason for the Pacing Stopped alarm has been resolved, that part of the alarm message is
removed from the display. The audio alarm continues. You must press the
resume pacing, remove the remainder of the alarm from the display and silence the audio alarm.
Pacing has stopped. There has
been a power failure during
Pacing has stopped. A pads off
condition has been detected
Pacing has stopped. The
HeartStart XL+ has detected an
error which prevents delivery of
Pacing has stopped. The Therapy
cable is disconnected from the
Therapy is disabled due to an
Pacing has stopped. The primary
ECG lead has become invalid in
Demand Mode pacing.
The actual delivered pace pulse
current is less than the selected
The battery's charge level is low.
20. Once generated, they appear as
on page 33.
Type of Alarm
Red Alarm message with
soft key to