2 DETT - Instructions for Use
IntelliVue Rev. G.0 or higher allows the configuration of either a second MIB/RS232 port or the
combination LAN interface and MIB/RS232 port for Data Export. The DtOut2 driver is used to
connect a second port to Data Export. Only one connection is able to request wave data at a time, the
other connection responds with a notification that wave polling is not possible.
Debug Information Window
If the debug mode for one or more detected and available patient monitors is selected debug details are
displayed in this window. Refer to chapter "the Menu" for further details on how to enable / disable
the debug mode.
The presentation of the data reflects the format described in the current revision of the Data Export
Within the debug information window the displayed strings can be selected by highlighting and copied
to a clipboard. This allows export of data into e.g. MS Excel, Word etc.
The displayed data can be deleted within the debug information window by highlighting it and using
the Windows delete feature. By pressing "CTRL +A" all information in the debug information window
field can be selected for further processing.