Other functions: the
1047358 Rev C
5. Reconnect the patient circuit to the mask/interface.
6. Select Start Ventilation to initiate ventilation.
If Test Failed is displayed, check for leaks in the patient circuit, and install an
exhalation device with lower leak characteristics. Repeat test. If the exhalation
port test fails again, the intentional leak is unknown and Tot.Leak rather than
Pt. Leak is displayed in the patient data window.
From the Menu window you can adjust user preferences.
Use Brightness to adjust the screen for optimum daytime or nighttime viewing.
Use Loudness to adjust the volume of the alarm and touchscreen audible
feedback. You will hear audible feedback as you go through the selections.
See "Selecting the mask and exhalation port" on page 6-7.
Vent Info (ventilator information)
The Ventilator Information window displays version and other information
specific to your ventilator.
Respironics V60 Ventilator User Manual