Notice: Read and understand all WARNING and CAUTION
statements listed in the safety section of this operation manual
prior to using this meter. Set the function select switch to the
OFF position when the meter is not in use.
1. ALWAYS turn the rotary function switch to the OFF position
when the meter is not in use
2. If "OL" appears in the display during a measurement, the value
exceeds the range you have selected. Change to a higher
DC/AC CURRENT MEASUREMENTS
WARNING: Disconnect the test leads from the meter before
making current clamp measurements.
Set the Function switch to the 400ADC,
40ADC, 400AAC or 40AAC range. If the
range of the measured current is not
known, select the higher range first then
move to the lower range if necessary.
For DC current measurement, press the
ZERO key to null the meter display.
Press the trigger to open jaw. Fully
enclose one conductor to be measured.
Note: Do not put more than one wire inside
the current clamp.
4. The clamp meter LCD will display the