Performance Verification Tests
These instructions describe how to test the pacing function. This test only
needs to be run if the Pacing option is installed.
If all results are as described, the unit passes the test. Return to the main
Diagnostic Test menu by pressing
If there is any failure, begin troubleshooting and repairing the unit as needed.
See "Troubleshooting" on page 3-1 and Table 3-10 "Operational Problems -
Pacing" on page 3-25.
Set up the test.
Disconnect the paddles and connect the pads cable. Connect the pads
cable to the Pacer tester.
From the Diagnostic Menu, select Pacer Test and press
display should look similar to Figure 2-6.
Figure 2-6 Pacer Test Display
The screen display of Pacer Status will change to Stopped.
begin at the default settings of 70 beats per minute (bpm) and 30 mA.
. The LED to the left of the button will illuminate.
. Pacer Status will change to Pacing, and pacing will
Performance Verification and Safety Tests