Install and use this appliance in accordance with the operation manual for safety and EMC (Electromag-
netic Compatibility). If it is not installed and used in accordance with the operation manual, it may cause
interference to other equipment and/or other risk.
To prevent fire or shock hazard, do not expose this appliance to rain or moisture.
This appliance must be earthed.
In Europe, use the AC power cord according to the recommendations as below, in order to comply with
EN60601-1 and EN60950-1.
Connect to the 230V receptacle of the room or the host equipment.
The AC power cord should be VDE approved and consist of core size 1mm
or bigger, length 2.5m or
shorter cord with IEC60320-1/C13 type, 250V 10A or higher rating connector and CEE(7)VII type, 250V
10A or higher rating plug.
Use the video cable, S-video cable, and/or the wired remote control according to the recommendations
as below, in order to comply with safety and EMC standards.
The video cable shall be 2 m long or shorter, 75Ω coaxial, 3C-2VT or equivalent one, with BNC plug at
The S-video cable shall be Mitsubishi Electric model JC-44S (1.5 m long or shorter, shielded wire) or
equivalent one, with DIN 4P plug at each end.
The wired remote control shall be Mitsubishi Electric parts No. 290P051010. (2m long or shorter,
shielded wire, with 3.5mm diameter stereo mini-plug and switch box.)
This product is to be employed with medical equipment, just for reference purpose, not for medical