GB/US: LED indicator
GB/US: Safety release mechanism
SP: Indicador LED
SP: Mecanismo de liberación de seguridad
PT: Indicador LED
PT: Mecanismo de liberação da segurança
Transmitter – User
SP: Botón de
PT: Botão de
GB/US: Neck cord
- Guía de Usuario
SP: Colgante para el cuello
PT: Cordão para o pescoço
Pulso Acionado a
MCT-212 AMBER with Wrist Band /
MCT-212 AMBER with Neck Strap / con correa para el
Código – Guia do
AMBER con Pulsera para Muñeca /
cuello / AMBER com cinta para o pescoço
AMBER com faixa para o pulso
3. GENERAL COMMENTS
The MCT-212 AMBER is a miniature waterproof transmitter,
designed for home health care signaling. In the event of an
Our wireless systems are very reliable and are tested to high
emergency, help can be summoned at the press of the
standards. However, due to their low transmitting power
(required by FCC, DTI and other regulatory authorities) there
are some limitations to be considered:
All MCT-212 AMBER units are supplied with a wrist band and
A. Receivers may be blocked by radio signals occurring on or
neck strap. They can be worn on the wrist or around the
near their operating frequencies, regardless of the code
The LED lights during a transmission to indicate a good battery.
B. A receiver can only respond to one transmitted signal at a
If the LED flashes during a transmission, this means that the
C. Wireless equipment should be tested regularly (at least
battery is low and must be replaced immediately. Anytime the
once a week) to determine if there are sources of
MCT-212 AMBER is pressed when the battery is low, a trouble
interference and to protect against faults.
signal will automatically be sent to the Base Unit.
To alert you of the trouble, a red LED will flash and a trouble
message will appear on the display of the Base Unit. All troubles
315, 433.92, 868 or other frequency
are reported to the Monitoring Center.
according to local requirements.
The transmitter case is hermetically sealed (no screws) to
Transmitter ID: Over 16 million possible code combinations.
ensure complete waterproofing of the case. Therefore battery
Minimum Transmit Duration: 2 sec
replacement must be performed by the factory only.
Power Supply: 3-volt Lithium cell type CR-2032
Battery Life Expectancy: 3 years (for typical use)
2. TEST, MAINTENANCE & ENROLLMENT
Good battery - LED lights steadily upon activation.
2.1 Enrolling Transmitters
Low battery - LED flashes upon activation.
In order for a transmitter to function, its ID must be enrolled
Note: If transmission is still possible despite the battery
into the Base Unit. The Transmitter ID is a 6 character code
condition, the MCT-212 AMBER will send a low battery report to
(ex: EB1772) that can be found both on the unit's package
and on the reverse side of the unit. To enroll a transmitter,
Operating Temperature: 0° to 50°C (32° to 122°F).
provide central station (either by telephone or by fax) of the
Dimensions: 39 x 37 x 13 mm (1-9/16 x 1-7/16 x 5/8 in.).
Weight: 25 g (0.9 oz).
2.2 Initial Test
This device complies with Part 15 of the FCC Rules and RSS-
Upon pressing the transmit button, the Base Unit will sound
and display simultaneously a message of calling for help. This
210 of Industry and Science Canada. Operation is subject to
the following two conditions: (1) This device may not cause
indicates that the transmitter is working properly.
harmful interference, and (2) this device must accept any
Note: If for any reason the MCT-212 AMBER does not result
interference received, including interference that may cause
in the above response from the Base Unit, please contact the
central station and verify the Transmitter ID located both on
WARNING: Changes or modifications to this unit not expressly
the unit's package and on the reverse side of the transmitter
approved by the party responsible for compliance could void the
beneath the wrist band.
user's authority to operate the equipment.
W.E.E.E. Product Recycling Declaration
For information regarding the recycling of this product you must contact the company from which you orignially purchased it. If you are discarding this product and not
returning it for repair then you must ensure that it is returned as identified by your supplier. This product is not to be thrown away with everyday waste.
Directive 2002/96/EC Waste Electrical and Electronic Equipment.