TABLE OF CONTENTS TABLE Of CONTENTS Before using the Monitor Introduction ... 3 Safety Information ... 4 Operating the Device ... 4 Care and Maintenance ... 5 Operating Instructions Before Taking a Measurement ... 6 Know Your Unit ... 7 Unit Display ...
Frequently measuring the blood pressure in your wrist will provide your physician with an accurate indication of changes in your blood pressure. The BP629 comes with the following components: • Monitor • Storage Case • 2 “AAA” Alkaline Batteries • Instruction Manual printed in English and Spanish • Quick Start Guide The BP629 IntelliSense Automatic Wrist Blood Pressure Monitor is intended for home use.
SAfETY INfORMATION To assure the correct use of the product, basic safety measures should always be followed including the warnings and cautions listed in this instruction manual. SAFETY SYMBOLS USED IN THIS INSTRUCTION MANUAL Indicates a potentially hazardous situation which, if not avoided, WARNING could result in death or serious injury.
SAfETY INfORMATION Do not use a cellular phone near the device. It may result in an operational failure. Use only Omron authorized parts and accessories. Parts and accessories not approved for use with the device may damage the unit. Use only 1.5 V “AAA” alkaline batteries with this device. Do not use other types of batteries.
BEfORE TAKING A MEASUREMENT To ensure a reliable reading follow these recommendations: 1. Avoid eating, drinking alcohol, smoking, exercising, and bathing for 30 minutes before taking a measurement. Rest for at least 15 minutes before taking the measurement. 2. Stress raises blood pressure. Avoid taking measurements during stressful times. 3.
UNIT dISPLAY Systolic blood pressure Diastolic blood pressure Memory symbol Average value symbol Irregular heartbeat symbol Pulse display Movement error symbol Battery low symbol Date/Time display Blood pressure level indicator Heartbeat symbol (Flashes during measurement) Deflation symbol...
IRREGULAR HEARTBEAT SYMBOL ( When the monitor detects an irregular rhythm two or more times during the measurement, the Irregular Heartbeat Symbol ( measurement values. An irregular heartbeat rhythm is defined as a rhythm that varies by less than 25% from the average rhythm or 25% more than the average rhythm detected while the monitor is measuring the systolic and diastolic blood pressure.
dISPLAY SYMBOLS This is a general guideline because blood pressure depends on age and morbidity. A lower home blood pressure goal may be recommended for certain patients, including diabetic patients, pregnant women, and patients with renal failure. WARNING Contact your physician for specific information about your blood pressure. Self-diagnosis and treatment using measured results may be dangerous. Follow the instructions of your physician or licensed healthcare provider.
BATTERY INSTALLATION CAUTION Use only 1.5V “AAA” alkaline batteries with this device. 1. Press the hook on the bottom of the battery cover and lift the cover off in the direction of the arrow. 2. Install 2 “AAA” size batteries so the + (positive) and - (negative) polarities match the polarities of the battery compartment as indicated. 3. Set the date and time. Refer to “Setting the Date and Time”.
BATTERY INSTALLATION BATTERY REPLACEMENT When the Low Battery Indicator appears on the display, turn the monitor off and remove all the batteries. Replace with 2 new batteries at the same time. Long-life alkaline batteries are recommended. WARNING If battery fluid should get in your eyes, immediately rinse with plenty of clean water. Contact a physician immediately. NOTES: • Turn the unit off before replacing the batteries. If the batteries are removed while the unit is still on, the date and time will be reset to that of the previous use.
SETTING THE dATE ANd TIME Set the monitor to the current date and time before taking a measurement for the first time. If the date and time has not been set, the measurement values are stored without any date or time information. 1. TO START THE SETTING Press the Date/Time Setting button ( ) to adjust the date and time.
SETTING THE dATE ANd TIME 3. SETTING THE MONTH Press the Date/Time Setting button ( ) to advance by increments of one month. Press the Date/Time Setting button ( ) to set the current month. The day flashes on the display. 4. SETTING THE DAY Press the Date/Time Setting button ( ) to advance by increments of one day.
APPLYING THE WRIST CUff CAUTION Read all of the information in the instruction manual and any other literature in the box before operating the unit. CAUTION This device is intended for use in measuring blood pressure and pulse rate in the adult population. Do not use this device on infants or persons who cannot express their intentions.
APPLYING THE WRIST CUff 5. Fold the remaining part of the wrist cuff back out of the way. APPLYING THE CUFF ON THE RIGHT WRIST When taking a measurement using the right wrist position the cuff as shown in the illustration.
TAKING A MEASUREMENT 1. Sit comfortably in a chair with your feet flat on the floor. 2. Hold your arm across your chest at heart level. Hold your elbow firmly to avoid body movement. Sit still and do not talk or move during the measurement. 3. Press the START/STOP button. All symbols appear on the display.
TAKING A MEASUREMENT The cuff starts to inflate automatically. As the cuff inflates, the monitor automatically determines your ideal inflation level. This monitor detects your blood pressure and pulse rate during inflation. The Heartbeat Symbol ( ) flashes at every heartbeat. Remain still and do not move until the entire measurement process is completed. NOTE: To stop the inflation or measurement, push the START/STOP button. The monitor will stop inflating, start deflating, and will turn off. 4. After the monitor has detected your blood pressure and pulse rate, the cuff automatically deflates. Your blood pressure and pulse rate are displayed. 5.
USING THE MEMORY fUNCTION Every time you complete a measurement the monitor stores the blood pressure and pulse rate in the memory. The monitor automatically stores up to 60 sets of measurement values (blood pressure and pulse rate). When 60 sets of measurement values are stored, the oldest record is deleted to save the most recent values.
USING THE MEMORY fUNCTION NOTES: • Press the MEM button repeatedly to display the next values. • Press and hold the MEM button to display the values faster. The date and time are alternately displayed with the measurement values. 3. Press the START/STOP button to turn the monitor off. TO DELETE ALL VALUES STORED IN THE MEMORY You cannot partially delete values stored in the memory. All values will be deleted. 1.
CARE ANd MAINTENANCE To keep your digital blood pressure monitor in the best condition and protect the unit from damage, follow the directions listed below: Do not forcefully bend the cuff. Do not fold tightly. Clean the monitor with a soft dry cloth.
ERROR INdICATORS SYMBOL Monitor could not detect pulse wave. Cuff under-inflated. Cuff not applied correctly. Cuff over-inflated. Device error. Movement during measurement. Irregular or weak pulses are detected. Batteries are worn. CAUSE Take another measurement and remain still until the measurement is complete.
TROUBLESHOOTING TIPS PROBLEM No power. No display appears on the unit. Measurement values appear too high or too low. CAUSES AND SOLUTIONS Replace both batteries with new ones at the same time. Check the battery installation for proper placement of the battery polarities. Blood pressure varies constantly.
fCC STATEMENT Note: POTENTIAL FOR RADIO/TELEVISION INTERFERENCE (for U.S.A. only) This product has been tested and found to comply with the limits for a Class B digital device, pursuant to part 15 of the FCC rules. These limits are designed to provide reasonable protection against harmful interference in a residential installation.
LIMITEd WARRANTY Your BP629 IntelliSense Automatic Wrist Blood Pressure Monitor, excluding the ® wrist cuff and batteries, is warranted to be free from defects in materials and workmanship appearing within 2 years from the date of purchase, when used in accordance with the instructions provided with the monitor.
Memory: Contents: NOTE: These specifications are subject to change without notice. SPECIfICATIONS BP629 (HEM-6200-Z) LCD Digital Display Pressure: 0 to 299 mmHg, Pulse: 40 to 180/min. Pressure: ±3mmHg or 2% of reading Pulse: ±5% of reading Automatic by electric pump...